Electrochemical fuel cell
Fuel cell sensor is unaffected by acetone, paint and glue fumes, foods, confections, methane, and practically any other substance found in the human breath.
Automatic after the subject blows for 4-6 seconds
±.005 BrAC up to 0.100 BrAC and ±5% above 0.100 BrAC, which meets DOT specifications
Within five seconds of sampling, depending on alcohol concentration
Immediate at zero alcohol reading, up to two minutes after a positive breath test
Instrument is immune to high levels of radio frequency interference.
Beeper signals fault conditions and changes in instrument status.
Two LED lights indicate status (Wait, Ready, Flow, Analyzing)
Dry gas or wet bath simulator.
Two-channel, RS-232 serial interface to printer or PC for test data transfer
Stores more than 100 test results including date and time. Reprints last breath test even if instrument has been switched off and main power source is removed. Stored tests can be downloaded to the printer or a PC.
Recommended Operating Temperature
23° to 104°F (-5° to 40°C). The fuel cell is automatically heated at low temperatures.
3¼” wide x 1¾” deep x 6¾” high (sensor) (8.3 cm wide x 4.4 cm deep x 17.1 cm high)
One pound (.45 kg)
Five “AA” batteries (included)
Included With Each Instrument
Printer, case, 3 rolls of printer paper, printer connector cable, 100 mouthpieces, operator’s manual
400 DH software for downloading of test results to a PC, rechargeable power unit, 12-volt power supply
One year, parts and labor